How to Prepare Gummy Samples for HPLC Analysis
- 50mL Centrifuge Tube
- 10mL Pipette Tips
- 0.5g Sodium Chloride - NaCL
- 2.0g Magnesium Sulfate Anhydrous MgSO4
- 10mLHPLC Grade Acetonitrile
- 10mL HPLC Grade Water
- 1g-1.5g Gummy Cannabinoid Sample
- Laboratory Spatula/Spoon
- Weigh Boats
- Centrifuge Test Tubes
- Analytical Balance
- 10mL Pipette
- Centrifuge Tubes
- Disposable gloves and safety glasses should be worn at all times when handling chemicals.
- Acetonitrile is severely irritating to the eyes and slightly irritating to the skin.
- Extended contact can lead to absorption and more irritation.
- Refer to Methanol MSDS in the event of skin or eye contact or inhalation.
Take a gummy sample and weigh it on the analytical balance. Record exact mass in mg.
Place 1 g to 1.5g of gummy in a falcon tube.
Use the 10mL pipette to transfer 10mL of HPLC grade water into the tube.
Cap and vortex for several minutes. Shape of the sample should begin to get lost.
Place in the sonicator at 45C for 15 minutes.
Vortex sample until it has dissolved.
Let the tube with sample equilibrate to room temperature.
Using a different pipette tip, Add 10mL of HPLC grade acetonitrile into the centrifuge tube.
Vortex sample for 1 minute until thoroughly mixed
Take the 0.5g bag of sodium chloride (NaCl) and add it to the centrifuge tube.
Take the 2.0g bag of magnesium sulfate (MgSO4) and add it to the centrifuge tube next.
Cap the centrifuge tube, vortex, and shake until salts are dissolved.
Place the centrifuge test tube in the centrifuge and turn on the centrifuge for 3 minutes.
Balance the centrifuge test tubes by placing each sample tube across from each other within the centrifuge. If only 1 sample is being prepared, counter-weigh the centrifuge by adding a 50mL tube containing enough water to be comparable to the volume of your sample.
How to balance samples within a centrifuge
Remove test tubes, take top layer and filter 1mL through 0.22um filter into a green vial
Gummy Sample Dilution for analysis with Cann-ID
- 1 mL disposable syringe
- 20ul-200ul pipette tips
- 0.22 um syringe tip filter
- 20ul-200ul Pipette
Place the small vials in the sample prep holding tray, from left to right starting with the green vial, followed by the yellow, and then blue.
- Use the 1 mL syringe to draw up 1 mL of extract solution containing sample
- Place a syringe filter on the tip of the syringe.
- Unscrew the green cap vial in the sample prep tray.
- Dispense liquid in the syringe into the open green cap vial.
Discard the syringe and syringe filter, and put the green cap back on.
Note: Every vial that contains methanol should be tightly capped whenever it is not immediately in use. The caps of the vials should be removed only when they are being used to transfer liquid in between the vials in use. This will prevent evaporation of methanol and provide you with more accurate results.
Vortex the vial briefly.
Set the micropipette to 40 uL and attach a clean tip.
Prepare/prime the pipette tip for transfer by drawing up and dispensing the green cap solution three times into the green cap vial.
Transfer 40 uL from the green cap vial to the yellow cap vial.
- Ensure that all of the solution has been transferred by drawing up and dispensing the yellow cap vial solution three times.
- Cap both vials and shake or vortex the yellow vial well.
- Discard the used pipette tip, attach a new, clean, one, and transfer 40 uL of the yellow vial solution to the blue cap vial.
- Once again, ensure complete transfer by drawing up and dispensing the blue cap vial solution three times.
Cap both vials and shake or vortex the blue vial well.
The sample is now prepared and ready for HPLC Analysis.
Documentation on Hazardous Chemicals
- Methanol MSDS
- Acetonitrile MSDS
- Sodium Chloride MSDS
- Magnesium Sulfate Anhydrous MSDS
How to Operate & Run Samples on Your Manual Injector HPLC
For instructions on how to warm up, shut down, and run samples on your HPLC, please read the attached PDF.
How to Calibrate your HPLC - Agilent 1220 Manual Injector
Calibrating the HPLC Purpose This document details how to properly calibrate an Agilent 1220 Manual Injection HPLC to deliver accurate results for cannabis potency testing. Responsibility It is the responsibility of the client to ensure that the ...
How to Change Mobile Phase Bottles on an HPLC
Changing Mobile Phase Bottles Purpose The purpose of this procedure is to correctly change mobile phase bottles on an HPLC. Scope This procedure applies to HPLC instruments used by Ionization Labs and its clients. Responsibility It is the ...
How to Calibrate your HPLC - Agilent 1220 Autosampler
Calibrating the HPLC Purpose This document details how to properly calibrate an Agilent 1220 Autosampler HPLC to deliver accurate results for cannabis potency testing. Responsibility It is the responsibility of the operator to ensure that the ...
Cann-ID Support Portal: Overview
Cann-ID Support Overview Customer Portal New User / Log In Portal URL: support.cannid.com If you have not already created an account, please set yourself up as a New User by selecting ‘New User? Sign Up’. You will be required to provide some very ...